Hey I replaced RAM u2460 and then re callibrated. I keep getting TEST 04 Fail 02. What does the 02 mean? Then I do exer 04 and get EOE2 5876 01 Aooo
It means the cal. data are corrupted. How to proceed: do a complete reaclibraion (dowonload service manual from www.ebaman.com for a 2465B or 2445B) If another recalibration fails, the scope can be abandoned.